STERILE CORD BLOOD COLLECTION UNIT (MSC128D)- anticoagulant citrate phosphate dextrose solution injection, solution
The MSC128D Sterile Cord Blood Collection Unit consists of 300 mL collection bag containing 21 mL of Anticoagulant Citrate Phosphate Dextrose Solution USP (CPD), a 40 mL rinsing bag containing 8 mL of CPD and two 12 gauge needles with a protective shield (SECUVAM) for the used needle.
Sterile, non-pyrogenic fluid path. Sterilized by steam.
This product is not made with natural rubber latex.
MSC128D - For collection of up to 200 mL of umbilical cord blood. Use aseptic technique.
If the collection volume is less than 60 mL, the cord blood should be processed within 24 hours of collection.
MSC128D Sterile Cord Blood Collection Unit contains:
Representative Product Drawing
II. COLLECTION PROCEDURE
Note: Umbilical venipunctures can be done before or after the expulsion of the placenta according to your local procedures.
3. Open clamps 3A and 3C.
4. Stimulate cord blood flow by massaging the cord from the top downward to increase volume of collection.
5. When collection is completed, close clamps 3A and 3C.
6. Remove needle 1A and slide down SECUVAM 2A (needle protector) over needle 1A.
7. If necessary, perform a second umbilical venipuncture on the upper part of the umbilical cord (near the mother)
a. Open clamps 3B and 3C.
b. Perform venipuncture on the umbilical cord using other needle 1B.
c. When collection is completed, close clamps 3B and 3C.
d. Remove needle 1B and slide down SECUVAM 2B (needle protector) over needle 1B.
III. RINSING OF COLLECTION LINE
IV. ADDITION OF ANTICOAGULANT INTO THE COLLECTION BAG
Use this step only for collection volumes greater than 150 mL.
V. END OF COLLECTION
a. 21 mL - if no additional CPD is added.
b. 29 mL - if additional 8 mL CPD from rinsing bag is added to the collection bag as directed in section IV.
4. For optimal cord blood quality, maintain the cord blood at an ambient temperature (18-26°C / 64.4-78.8°F) or cold temperature (4-12°C / 39.2-53.6°F).
VI. CONTACT INFORMATION
Manufactured for United States of America (U.S.A.) and Canada by Maco Productions Polonia, Sp. z o.o.
ul. Szwajcarska 22, 54-405 Wroclaw, POLAND
+48 71 37 60 110
For consumer information contact:
3075 Breckinridge Boulevard, Suite 405, Duluth, GA 30096, USA
1200 McGill College, Suite 1715, Montreal (Quebec) H3B 4G7
|STERILE CORD BLOOD COLLECTION UNIT (MSC128D)
anticoagulant citrate phosphate dextrose solution injection, solution
|Labeler - Maco Productions (265492868)|
|Registrant - Maco Productions (265492868)|
|Maco Productions Polonia sp zoo||367387284||manufacture(14498-004)|